Agilent 34941A Driver
Keysight / Agilent A Quad 1x4 50 Ohm 3 GHz Multiplexer In stock. EquipNet Listing #, Price: Best Offer, Manufacturer: Agilent Technologies,Model: A. Restricted Rights Legend Agilent AA RF Multiplexer Modules User's Guide Additional Safety Notices The following general safety precautions must.
|Supported systems:||Windows XP (32/64-bit), Windows Vista, Windows 7, Windows 8.1, Windows 10|
|Price:||Free* [*Free Registration Required]|
Agilent 34941A Driver
Used Electronic Testing and Measurement Equipment
Further, to the maximum extent permitted by applicable law, Agilent disclaims all warranties, either express or implied, with regard to this manual and any information contained herein, including but not limited to the implied warranties Agilent 34941A merchantability and fitness for a particular purpose. Agilent shall not be liable for errors or for incidental or consequential damages in connection with the furnishing, use, or performance of this document or of any information contained herein.
Should Agilent and the user have a separate written agreement with warranty terms covering the material in this document that conflict with these terms, the warranty terms in the separate agreement shall control. Software Revision Technology Licenses This guide is valid for the firmware that was installed in the instrument at the time of manufacture. However, upgrading the firmware may add or change product features. For the latest firmware and documentation, go to the product page at: Government Restricted Rights.
Software and Agilent 34941A data rights granted Agilent 34941A the federal government include only those rights customarily provided to end user customers.
Agilent provides this customary commercial license in Software and technical data pursuant to FAR It calls attention to an operating procedure, practice, or the like that, if not correctly performed or Agilent 34941A to, could result in damage to the product or loss of important data. It calls attention to an operating procedure, practice, or the like that, if not correctly performed or adhered to, could result in personal injury or death.
Failure to comply with these precautions or with specific warnings or Agilent 34941A elsewhere in this manual violates safety standards of design, manufacture, and intended use of the instrument.
General Do not use this products in any manner not specified by the manufacturer. The protective features of this product may be impaired if it is used in a Agilent 34941A not specified in the operation instructions. Do Not Modify the Instrument Agilent 34941A not install substitute parts or perform any unauthorized modification to the product.
Return the product to an Agilent Sales and Service Office Agilent 34941A service and repair to ensure that safety features are maintained. In Case of Damage Instruments that appear damaged or defective should be made inoperative and secured against unintended operation Agilent 34941A they can be repaired by qualified service personnel.
Make all connections Agilent 34941A the unit before applying power. Frame or chassis terminal Ground the Instrument This product is provided with protective earth terminals. To minimize shock hazard, the instrument must be connected to the ac power mains through a grounded Agilent 34941A cable, with the ground wire firmly connected to an electrical ground safety ground at the power outlet.
Any interruption of the protective grounding conductor or disconnection of the protective earth terminal will cause a potential shock hazard that could result in personal injury. Unit is not completely disconnected from Agilent 34941A mains when switch is off Caution, risk of electric shock Caution, refer to accompanying description Do Not Operate in an Explosive Atmosphere Do not operate the instrument in the presence of flammable gases or fumes.
Product Agilent 34941A To return unwanted products, contact your local Agilent office, or go to www.
Keysight-Agilent A - General Equipment Other Types of Equipment
Technical Support If you have questions about your shipment, or if you need information about warranty, service, or technical support, contact Agilent Technologies: In the United States: You can also contact your Agilent Technologies Representative. Do Not Remove the Instrument Cover Only qualified, service-trained personal who are aware of the hazards involved should remove instrument covers. Always disconnect the power cable Agilent 34941A any external circuits before Agilent 34941A the instrument cover.
That DoC applies to the A mainframe and all available plug- in modules.
Agilent AA RF Multiplexer Modules **
Both modules contain four banks of latching switches. Each bank consists of three Form C relays see the simplified schematic on page 4.
To create a larger switching Agilent 34941A, you can connect the banks in this module and connect to banks in other RF MUX modules. The important differences between the two RF MUX modules lie in their characteristic impedance and their use of connectors external cables are not provided with the module: When installing SMA cables on the A module, it is recommend that you tighten them to 0. To help prevent damage and contamination, do not remove a connector's protective Agilent 34941A until immediately prior to installing a cable on that connector.
Because of the space constraints between connectors on this module, right-angle plugs and SMB adapters are not recommended. Operating Considerations. That guidance sets maximum per channel current and power ratings at rated voltage for pollution degree 1 dry and pollution degree 2 possible condensation conditions, for each Agilent 34941A the GP Agilent 34941A.
Instead, all signal connections are made through the visible connectors via external cables.